|usage||measurement of total volume of the liquid passed through, instantaneous flow rate and pressure|
|the principle of the measurement||transit-time|
|type||- one beam (1B)|
- remote and compact version
|rated piping size||DN32 to DN300|
|maximum permitted pressure||1.6 MPa, 4 MPa (classes MAP 16 and MAP 40)|
|measurement accuracy||according to EN 14154|
|flow sampling period||1 sec|
|tempetarure of liquid||up to 150°C|
|protection class||IP67 / IP68 in remote version|
|power source||Li battery 3.6V/16Ah, garanted battery lifetime 6 to 8 years|
Ultrasonic flow meters FLOMIC FL103X
Ultrasonic flow meters of the type series FLOMIC FL103X (FL1034 - the compact version, FL1035 - with remote electronic unit) are intended for measurement and storage of the measured data of instantaneous flow rate, total volume passed through the meter sensor, of cold and hot water and other liquids of similar properties in a wide range of industrial applications. Optionally, the meters can be provided with pressure gauges to monitor the fluid pressure in the piping.
The FLOMIC FL103X meters are noted for high measurement accuracy and long-term stability over a wide range of measured values. They have equipped by internal battery and they can be installed anywhere without availability of external power. The meters include no movable parts and, compared to conventional flow meters, their hydraulic losses are significantly lower.
The associated electronic unit provides for processing of the sensor signals and visualisation of the values of instantaneous flow rate, total fluid volume passed through the meter sensor and fluid pressure, as well as for archiving of the measured data in specified time intervals in the internal meter memory unit. The archived data can be retrieved either via an optical interface, RS 232 communication line or GSM module for further processing by, for example, a notebook computer. On-line processing of the measured values is also possible using the pulse or current (4 to 20mA) outputs.
Posted by: rudy wiratama
wiratama Updated at: 23.22